Developer Terms and Conditions › The Development › Hardware › Transmittance mode for liquid samples › Reply To: Transmittance mode for liquid samples
I have an application which requires measuring sugar content such as the brix scale during fermentation. I had been communicating with Miriam regarding the liquid attachment, but she has just come back and said there are problems, so it will not be available now. I was going to use this as a starting point, but the end application will require the SCiO to be enclosed in a liquid proof vessel and emersed in the liquid. Miriam recommended glass over plastic. We are fine with fabricating the vessel ourselves, but I am trying to get any help possible on how far the SCiO needs to sit from the outer transparent wall and if it needs any special focusing, and any other thoughts you may have.